DSO 2150 USB DRIVER DETAILS:
|File Size:||13.4 MB|
|Supported systems:||Windows 10, Windows 8.1, Windows 8, Windows 7, Windows 2008, Windows Vista|
|Price:||Free* (*Free Registration Required)|
DSO 2150 USB DRIVER
Hantek 3, HP OWON RIGOL Tektronix 4, Whatta heck. Does anybody know how to adjust the Vpp so i can get a nice 2Vpp from the test terminal? Did it at dso 2150 usb kHz change as bescribed in the atteched probe-manual changing form?
I really don't want you to run in cycles, but 1kHz is really something other than kHz when it comes to calibration! Check the vertical calibration with DC voltages, eg. dso 2150 usb
Use dso 2150 usb voltage divider and compare the reading of the scope with a digital multimeter. As far as probe calibration goes, with the 1kHz test signaladjust for proper shape of the square wave. Do not adjust at any other frequency as you will not get dso 2150 usb expected results, which you have mentioned. Quote: rotating the screw only changes the peak to peak voltage, not the form of the signal Obviously you did the above at a much higher frequency.
Dso 2150 usb you have adjusted the probes, you will not need to do it again. Just do a quick check every week or so; that should be more than enough. Waveform is almost square! F is a lot less then calculated, but i guess the formula is very very approximate.
Agaim, signal is alsmost square! So, the distorted signal was due to bad R and C combination for this chip. Most likely it overloaded the chip.
|imation 40x24x40 cd burner||Other Useful Business Software|
|tvs rp 3150||Search form|
|hp laser printer 1150||Shop All Categories|
|asus pro31j||Ask a Question|
I have calibrated the probe at 1KHZ to see the square signal. Yes, at 1Khz when i rotate the crew it change the form of the signal as it is described in the manual for the Dso 2150 usb.
I have calibrated it at 1Khz and then did the all measurements again. With pF cap i still get the distorted signal.
- DSO USB Download - Portable PC-baseddigital oscilloscope
- Couple of question about HanTek DSO oscilloscope AVR Freaks
- DIY Audio Projects Forum
- That's not you, right?
With 22pF Cap i get what i described above. Quote: Check the vertical calibration with DC voltages, eg. I have checked the test signal with multimeted.
So, the test signal is fine. Here's calibration signals on dso 2150 usb DSO, along with a vcc and clock power up shot from some other project.
Hantek DSO 60MHZ MSa/s USB Oscilloscope (HCTEST) -
CLK is running at 3. Dso 2150 usb do a zero-calibration, make sure the scope has been on for a few dso 2150 usb, so it had time to warm up, then go to Util menu, Calibration, and Zero-Calibration.Make your computer a powerful digital storage oscilloscope capable of FFT calculation with the affordable 60MHz Hantek DSO PC based USB. PC USB Oscilloscope Current Location：Home > Product Details>DSO Product Details One computer can connect many DSO, extend channel easily.